Please rate topic.
Average 3.9 of 61 Ratings
Thank you for rating! Please vote below and help us build the most advanced adaptive learning platform in medicine
When comparing unaged, radiated, cross-linked, ultra-high molecular weight polyethylenes (XPE) treated with remelting or annealing, annealed XPE has
fewer free radicals, better mechanical characteristics, and lower wear than remelted XPE.
fewer free radicals, worse mechanical characteristics, and lower wear than remelted XPE.
more free radicals, better mechanical characteristics, and equivalent wear than remelted XPE.
more free radicals, worse mechanical characteristics, and equivalent wear than remelted XPE.
equivalent free radicals, better mechanical characteristics, and lower wear than remelted XPE.
Select Answer to see Preferred Response
A 75-year-old patient returns for follow-up after undergoing bilateral total hip arthroplasty (THA). The right hip is a hybrid THA performed 12 years ago, whereas the left hip is a cementless THA performed 10 years ago. Both acetabular components are the same type, same size, and from the same manufacturer. Both femoral heads are 28-mm cobalt-chromium components. What is the most likely explanation for the advanced polyethylene wear in one hip?
Presence of a third-party abrasive particle
Disproportionate use of the limbs by the patient
Defective femoral head
Method of fixation of the femoral component
Method of sterilization and shelf life of the polyethylene
When polyethylene is exposed to radiation and subsequently heated, certain chemical changes occur in the material. Which of the following statements best describes these changes?
The process converts an otherwise interpenetrating networking structure of polymer chains into a linear, high molecular weight polyethylene macromolecule.
The process increases the ductility of the material.
The process leads to fewer particles that are larger in size than the untreated material.
The process improves (lowers) the wear rate but may increase the risk of fracture.
The process decreases the wear rate of the material, compared to untreated polyethylene, when tested against a rough counterface.
Sterilization of ultra-high molecular weight polyethylene by gamma irradiation in air will degrade its wear performance because of
Which of the following manufacturing techniques of UHMWPE results in the lowest susceptibility to fatigue crack formation and propagation in joint arthroplasty bearings?
Ram bar extrusion with secondary machining into the desired product
Hot isostatic pressing into bars with secondary machining
Irradiation with 10 Mrad of radiation achieiving a polyethylene crystallinity of >99%
Direct compression molding from PE powder to the desired product
Addition of calcium stearate to the polyethylene resin followed by compression molding into bars with secondary machining into the desired product
The polyethylene cup shown in Figure A has been sterilized in packaging that contains oxygen. Which of the following statements is true about the process of polyethylene sterilization?
Irradiation of polyethylene in argon quenches free radicals
Polyethylene sterilization with ethylene oxide causes cross-linking of the polymer and minimizes oxidation following implantation
Oxidation occurs following polyethylene implantation regardless of sterilization process
Irradiation of polyethylene in nitrogen quenches free radicals
Polyethylene sterilization with gas plasma causes cross-linking of the polymer and minimizes oxidation following implantation
In evaluating methods of polyetheylene sterilization for hip arthroplasty, gamma-irradiation in air compared to irradiation in an inert substance results in which of the following?
No difference in regards to outcome
Higher rate of cross-linking when irradiated in air
Lower rate of oxidation when irradiated in air
Accelerated wear and failure when irradiated in air
Better wear resistance and longevity when irradiated in air
A 55-year-old male undergoes a revision total knee arthroplasty of an implant that is only 3 years old. At the time of surgery, the tibial polyethylene liner shows catastrophic delamination and cracking. What is the most likely cause of this extensive, accelerated wear of the polyethylene liner?
Sterilization in ethylene oxide
Gamma irradiation of the polyethylene liner in the presence of air
Gamma irradiation of the polyethylene liner with vacuum packaging
Gamma irradiation of the polyethylene liner in nitrogen
Gamma irradiation of the polyethylene liner in argon
Polyethylene gamma irradiation in nitrogen gas results in which of the following?
What polyethylene processing step results in increased polyethylene wear and subsequent osteolysis?
Gamma irradiation in air
Remelting after gamma irradiation in nitrogen
Ethylene oxide sterilization
How are free radicals removed from highly cross-linked polyethylene?